H. Pylori Antibody Rapid Test Kit (Lateral chromatography) is a Lateral chromatography intended for the rapid, qualitative detection of IgG antibodies specific to Helicobacter pylori in human serum, plasma, whole blood or fingertip whole blood as an aid in the diagnosis of H. pylori infection in patients with clinical signs and symptoms of gastrointestinal disease. The test is only to be used by medical professionals.
The kit is immunochromatographic and uses capture method to detect H. Pylori Antibody. H. Pylori antigens are bond at the Test line (T). When the sample is added, IT will form complexes with H. pylori antibodies in the samples, and microsphere-labeled mouse anti-human igg antibodies bind to the complex at the T lines to form visually visible lines. If there are no anti-H. Pylori antibodies in sample, no red line is formed in the Test line (T). A built-in control line will always appear in the Control line (C) when the test has performed properly, regardless of the presence or absence of anti-H. pylori antibodies in the specimen.
|Test Cassette||1 test||25 tests|
|Sample Diluent||1 bottle||25 bottles|
|Dropper||1 piece||25 pcs|
|Alcohol Pad||1 piece||25 pcs|
|Disposable lancet||1 piece||1 piece|
Step 1: Sampling
Collect human Serum/Plasma/Whole blood properly.
Step 2: Testing
1.Remove an extraction tube from the kit and a test box from the film bag by tearing the notch. Put them on the horizontal plane.
2.Open the inspection card aluminum foil bag. Remove the test card and place it horizontally on a table.
3. Use a disposable pipette, transfer 10μL serum /or 10μL plasma/ or 20μL whole blood into the sample well on the test cassette. Start counting.
Step 3: Reading
10 mins later, read the results visually. (Note: do NOT read the results after 15 mins!)
Colored bands appear at both test line (T) and control line (C). It indicates a positive result for the detection of H. pylori-specific IgG antibodies.
Colored band appear at control line (C) only. It indicates the absence of H.pylori-specific IgG antibodies.
No visible colored band appears at control line after performing the test. Insufficient sample volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the test procedure and repeat the test using a new test device.
|Product Name||Cat. No||Size||Specimen||Shelf Life||Trans. & Sto. Temp.|
|H. Pylori Antibody Rapid test kit (Lateral chromatography)||B011C-01||1test/kit||Serum/Plasma/Whole Blood||18 Months||2-30℃ / 36-86℉|